Radiography medical 200ma chest x ray machine
1. Usage:General radiography, Filter-ray device radiography.
The unit can used for general radiography, Bucky radiography of body's head, chest, foot and hand, bones in lying.
2. Unit Configuration:
Control, HT generator unit, Ht cable, X-ray tube unit, Diagnostic table, Collimator
3. Main characteristics:
2. Single-table single tube.
3. Single-phase, full wave rectification.
4. Stationary anode X-ray tube assembly.
5. X-ray tube filament s voltage regulator, space charge compensator.
6. Radiograph plain table wimple structure and easy assembling.
MAIN TECHNICAL INDEXES:
|Demand of power supply||Capacity||Two-phase 30KVA|
|Voltage Frequency||380V/220V ± 10%50HZ ± 0.5Hz|
|Internal resistance||0.75Ω (380V)0.25Ω (220V)|
|power supply||Adjustable range||380± 10% at220V being adjustable continuously|
|Radiograph||Voltage||50-100KV being adjustable continuously|
|current||Small focus: 50mA100mA, Big focus: 50Ma 100mA 200mA 300mA|
|time||0.04~6.35s 23grades together grading a according to R10 coefficient|
|Max DC output voltage||100KV|
|Max DC output current||200mA|
|Focus||Small focus 1× 1mm big focus2× 2mm|
|Radiographic table||Length*width*height of bed surface||2000mm× 665mm700mm|
|Moving range of bed surface||640mmin Length, 200mmin breadth|
|Ray-filter of table||Travel in length ≥ 500mm|
|Grid density N=28|
|Convergence distance f=100cm|
|Grid ratio r=8|
|Column of X-ray tube unit||Moving Longitudinally along Radiographic table||1000mmor 1800mm|
|Distance to friend as moving up and down (along column)||650~1800mm|
|Rotary around the center of cross arm||± 180°|
|Rotary around the shaft line of X-ray annular tubes||-10° ~60° ~ +120°|
|Max size of cassette||356mm× 432mm(14× 17)|